Environmet Friendly Fiber Optic Dental Drill Equipment Push Button
NSK Fiber Optic Handpiece
6-Hole Fiber Optic Handpiece With Quick Couplin
Chuck Type:Push Button, Compatible with NSK Machlite/Phatelus
Spray:Triple water spray
Working Air Pressure:0.20-0.25MPa(0.22MPa prefrred)
Rotation Speed:350,000-400,000R.P.M(Standard Head);
Dental Handpiece Application:
- It has unique appearance design and fluent outline.
- Smooth as silk, it has excellent texture.
- With people-friendly design, easy for operation.
- With best quality cartridge, easier for cleaning.
- With high-speed turbine rotation, it has better performances.
- Chucking power of chucking needle is 20N-45N.
- It is noise-free and stable.
- Applicable Bur: 1.6mm and should be conformed to ISO standard.
1. Clamping force of handpiece is:20-45N
2. Noise not exceed 70dB
3. At 25-30 atmosphele pressure,350,000-400,000R.P.M(Standard
4. Optical handpiece ,3 point water spray,double air tracks, quick
coupling, light guide vitreous, illumination≥25000LUX. Voltage is
3.3V±0.1V, LED light.
Environmet-friendly fiber opitic handpiece tube:
If the dental unit don’t have the optical fiber system,please
insert our optical fiber tubing and electricity borad in the dental
then it can use our optical fiber handpiece and inclube the high
speed handpiece which compliance with international standards.
1. Power in:12-24VFIBER OPTIC HANDPIECE
2. Power out:2.5-3.5V(adjustable)FIBER OPTIC HANDPIECE
3. Output delay:0-12s(adjustable)
4. Linkage air pressure:P＞0.2Mpa
5. Control ways:2 ways
1. Considering safety first, please be careful to use it with full
2. Be limited to dentists who held the dental treatment
3. Strictly forbid decomposition and reconstruction by
4. Please run point testing on outside of patients oral beforehand.
If feel anomalies, please stop using immediately and contact with
our after-sales center.
5. When appears abnormal during using, please stop it immediately
and contact with our after-sales center.
6. It will cause a high noise using a center-vibration or bended
bur and a handpiece of which bearing has been worn, A long time in
this state will be harm to people’s hearing. Forbid using a
vibration or bended bur, if appear high noise, please replace the
shaft core early
7. Do not use a bending , cracking, deformation and unqualified
bur. If using it, may cause a snap, or a flying off to injury
accident during operation process.
8. Please ensure the bur chuck be clean. If with contaminant inside
of it, may causes bur clamping weaken or shaft core vibration.
9. Please run the bur under the provision rotation-speed by bur
10. Do not crash, especially to avoid falling.
11. Please install a bur or a standard stick before using.
12. Aslo remember to install a bur or a standard stick when not in
13. Please unload the bur and handpiece when the running stopped in