Stainless Bearng High Speed Dental Handpiece 0.20MPa - 0.25MPa
Working Air Pressure
1. Push Button High Speed Handpiece M4/B2
2. Working Air Pressure:0.20-0.25MPa(0.22MPa prefrred)
3. Rotation Speed:350,000-400,000R.P.M(Standard Head);
4. Bur Applicable:Φ1.595-1.600mm"
5. Spray:Triple water spray
Consists of handpiece head, handle, caudal thread, cartridge.
1. Air pressure 0.25~0.30mpa,must be filtrated.
2. Use high speed dental bur with shaft diameter
Φ1.59~1.60mm×21-23mm and roghtness less than 0.8um.
3. The amount of handpieces spray or water spray should be able to
4. Hard glass fiber,can be repeatedly sterilized in high
temperautre and pressure.
Only used for drilling, molar of stomatology clamping high-speed
HPA009 Handpiecewe Provide Two Kind Of Head For Choice
1. Clamping force of handpiece is:20-45N
2. Noise not exceed 70dB
3. At 25-30 atmosphele pressure,350,000-400,000R.P.M(Standard
1. Considering safety first, please be careful to use it with full
2. Be limited to dentists who held the dental treatment
3. Strictly forbid decomposition and reconstruction by
4. Please run point testing on outside of patients oral beforehand.
If feel anomalies, please stop using immediately and contact with
our after-sales center.
5. When appears abnormal during using, please stop it immediately
and contact with our after-sales center.